Packing Details: 650*495*350mm, 0.05CBM , NW:8KG, GW:10KG
Elisa Reader, 8 channels optical system, 5 secs/96 wells(single wavelength)-some are 6 secs, 10 secs /96
wells ,.7 secs/96wells(double wavelength )-some products are 20 s , 4 filters.
Measurement System : 8-Channel optical system
Measurement Range : 0.000-4.000Abs
Photometric Accuracy : ±1% or ±0.001Abs
Repeatability : <±0.3%
Stability : <0.002Abs
Linearity : <±0.5%
Reading Speed : <5secs/96holes(single wavelength),
Filters : Standard: 4 filters (405, 450, 492,630 nm) or other wavelength on request..Totally
10 filter can be installed.
Full One year Warranty
The Elisa9602G Microplate Reader is a semi automated bench top instrument focused on clinical immunity-test in
hospitals or epidemic preventing stations. It has good characters such as high sensitivity, high accuracy, high speed of
detection and the easiness of operation etc. Display is a liquid crystal one with alphanumeric and graphic options.
Parameters can be checked on the display without using difficult command sequences.
II Main Functions
The instrument is compatible with registered diagnostic EIA kit, primarily used in hospitals or some detection related to
III Function and Technical Specifications
1 Fundamental parameter
1) Range of working wavelength :400nm-800nm;
2) Range of detection:0.000A -4.000A;
3) Resolving power: 0.001 A;
4) Filters when leaving factory: 405nm, 450nm, 492nm, 630 nm, with the maximum array of 10 filters between
5) Reading speed: single wave 5s, dual wave 10s;
6) Dimension: 420×325×210 mm(W*D*H)
7) Weight: 10 kg
2 Technical Specifications
1)Characters of the filters
Table 1: Characters of the Filters
2) The errors of absorbance accuracy are as follows:
Table 2: Absorbance Accuracy
Absorbance (A) Accuracy Error
3)The Linearity Error are as follows:
Table 3: Linearity of the Instrument
4) Reproducibility Error: ≤0.5%;
5) Stability Error: ≤±0.005A/10min;
3 Main functions
1) Provided with power on self-checking. Error data will display on LCD monitor if something wrong with the photoelectric
2)Filter choosing-mode: Automatically
3) Checking mode: Single wave/dual wave detection; single aperture /dual aperture detection
4) Input mode:Key in touch screen;
5) Input format: Impute capital letter/lowercase letter/number at random;
6)Item parameters: Provided with function of set/save/ inquiry 120 detection parameters, this is called in later usage.
7)Defection of calibration: with the function of calibration-defection,which can save/modify 120 parameters of calibration
8)Detecting method:Absorbance/qualitative/quantitative detection;
Qualitative test: single critical value checking; dual-critical value ,threshold level testing, gray-area setting;
Quantitative test : single point ,parabolic interpolation, linear regression, semi-logarithm regression, logarithm regression,
logit-log , enzyme kinetics, PCR testing;
9)Saving function:Save the checking results of 100 pieces of plates to be called in later usage.
Absorbance: display/printout originally absorbable value;
Qualitative checking: Display/printout absorbable value, critical value, S/CO, qualitative results;
Quantitative checking: Display/printout absorbable value, density, standard reference value,quantitative results.
11) QC function: Inquiry/printout the QC data and chart;
12) Communication: Equipment with RS-232 interface.
13) With self-monitoring function(real-time)on each testing light-source signal channel.
14) With dormancy function on light-source lamp.
|Reading Domain||850 nm|
ELISA microplate reader
1. Stand-alone microprocessor controlled
photometer designed to measure absorbance in 96 well (12X8)
2. Programmable reading between 1 to 96 wells Yes
3. Reading time: single wave 5s/96 plate, dual wave
4. Reading Domain: up to 4.0 OD in range 400 -
5. Dynamic range: 0 to 4.000 OD Yes
6. Linearity: < ±0.5% from 0 - 3.500 OD Yes
7. Precision: 0 to 3.500 OD: ±1% or ±0.001 Abs
8. Wavelength range: 400-800 nm Yes
9. Light source: Tungsten halogen lamp Yes
10. Filters: up to 6 changeable filters (on board to
be delivered with at least 450 nm, 490 nm, 620 nm)
11. Capacity: 120 assay procedure, 100 plates, and
20 standard curve data (for quantitative determination)
12. Result interpretation: Positive, negative, doubtful Yes
13. Plate validation: up to 6different criteria,
possibility of exclusion of nonconforming values and
reinterpretation of the results
14. Results printing: exactly data, and
15. Self test Yes
16. LCD display Yes
17. Power: 220-240 V 50/60 Hz Yes
18. Printer interface: (to be delivered with printer) With Inner heat printer or external printer,all is ok
19. Computer interface Yes RS232 interface
20 .If you need, we also can supply Laser, printer Yes ( We can give a Laser printer for this machine
20.1.Compatible with "ELISA Microplate automated
20.2. A4 paper format
20.3. Spare cartridge -1 Yes
20.4. Printing speed not less then 3 ppm Yes
20.5. Compatible MS Windows Yes
20.6. - Workload - 1000 pages per month, Printer
resource - 20 000 hours
20.7. Power source -220 v Yes
1. The installation is so simple that operators can install it by themselves according to the user manual.
2. HEPO INTERNATIONAL will provide the customers with free training service in our factory.The traffic and
accommodation charge will be on the customers' account.
Also, HEPO INTERNATIONAL will provide free technical support via e-mail, fax or telephone at any moment.
3. Full warranty - 1 years
4. Technical documentation in English Full One year Warranty